Vehicle non -metallic material atomization test meter is a testing instrument for the interior materials of the automotive industry in Shanghai QINSUN. Three test methods: Misty Method and Weight Method.
Vehicle non -metallic material atomization test meter is suitable for volatile organic substances in automotive interior materials, such as PVC, textiles, leather, non -woven cloth, rubber plastic products and other materials At high temperature, the evaluation of the performance of the volatile ingredients can also be used for the determination of the high temperature atomization phenomenon of the high temperature light of the front of the car.
Precautions for the use of non -metallic material atomization test meters for vehicles:
1) Daily precautions:
1. The operator carefully read the manual of this instrument to master various operation requirements.
2. Keep the machine clean, and the non -test machine time should cover the cover on the machine and cover the dustproof cloth on the whole machine.
3. On the basis of familiarity with the principle of this operation manual and electrical appliances, the operator may be eliminated by the device conventional maintenance and failure, and the customer is strictly prohibited from dismantling the machine without authorization.
4. The use of temperature control device should be carried out strictly in accordance with the operation instructions, otherwise the responsibility should be on the responsibility.
5. The sink of the temperature control device needs to be added to the sink. It is recommended to replace the distilled water every two months.
2) Special precautions:
1. The temperature of the hot oil in the host is heated to 150 ° C, and it is strictly forbidden to heat it above 150 ° C.
2. The indoor environmental temperature of the device should be at least 0 ° C, because the heat loading oil becomes thick as the temperature decreases, affecting heat transfer. The distilled water in the temperature control device will also freeze at 0 ° C and cannot circulate.
3. When the sink of the temperature control device has distilled water, its temperature can only be set at 5 ° C ~ 80 ° C to avoid boiling and freezing.
4. In the case where the host and the temperature control device are not liquid -free, do not start the device, otherwise there is a danger of burning equipment.
5. When the temperature of the liquid medium in the groove is higher than 45 ° C, do not turn on the cooling button.
6. The final measurement result of this test instrument is the luster of the glass plate and the weight of the aluminum foil, so it is strictly forbidden to touch the surface of the glass plate that has been touched or touched with aluminum foil.
7. The cleaning process of the attachment must be carried out strictly in accordance with the operation steps to avoid affecting the test results.
8. Glass after the testRiders and aluminum foil are strictly prohibited from contacting the sun and blowing off the hair.