Terraper test machine structure and size:
The structure and size of the tensile test machine should be suitable for test requirements for the test material, and its performance does not have an adverse effects on the test results.
The movement distance of the mobile clip should fully meet the needs of the maximum extension of the sample in the stretch test. In the case of large extension of the test materials, the movement distance may exceed 1m.
Tension test machine > Sample sandwiches:Dumbbell -shaped, parallel strips, and stretching samples similar to shaped. The test machine should be equipped with the automatic clamping type with the tensile force, and can generate uniform pressure along the entire width of the sample (such as a wedge -shaped chip or Pneumatic chip). For testing hard materials, the piping of the spiral clamping force is also applicable. The method of holding the sample should be as possible to prevent the sample from slipping from the pinch.
For the test ring sample, the test machine should be equipped with two rolled wheels that can be rotated freely, at least one of them can use the test machine with a speed of 3r/min ~ 50r/min. Rotate freely so that the ring -shaped sample of the subject is uniform during the test. The diameter of the corresponding large ring -shaped sample (inner diameter 44. 6 mm) roller should be 25 mm; the diameter of the corresponding small ring sample (8.0 mm inner diameter 8.0 mm) should be 4.5 mm.
For the adhesive intensity test using the stripping method, the test machine should be equipped with a pinch described in the relevant test method, or the entire width of the sample should be evenly clamped evenly. Powerful chip.
The method of holding the sample should prevent the sample from slipping relative to the pinch. When the bonding sample is made of different materials, the corresponding chip corresponding to each adhesive may need to be designed.
Touch screen desktop type Landario test machine > is a simple tension test machine. The structure is simple and convenient to operate. It adopts a touch -screen control system, rotates through the drive motor, and then drives the roller screw of the driver to rises and decreases after the transmission mechanical mechanism is deceived to complete the stretching or compression test of the sample. The force value is output by the sensor, the feedback display, the test speed and force valueThe change curve can be displayed in real time.
It is simple and convenient to operate, especially suitable for testing instruments to control the quality of product quality on the production line. This machine can be equipped with various fixtures to achieve different test needs, suitable for textiles, film, electronics, metals, plastic, rubber, textiles, synthetic chemical products, wire and cables, leather and other industries. >Specification parameters:
1, capacity: within 1-200kg Self-selected > 2, accuracy level: The display ± 0.5%(5%-100%of the full range) level 0.5
3, the resolution : 1/50000
4. Power system: Step motor+drive5. Control system: TM2101-5-inch color touch screen control (Available computer function)
6. Data sampling frequency: 200Times/SEC
7. Itinerary: 600mm
8. Test width: about 100mm
9. Fast range: 1 ～ 500mm/min
10 , Safety device: overload, emergency shutdown device, up and down stroke limited device, leakage protection device
11. Printer: Automatic printing form (Chinese) Point value, breakpoint ratio, date)
12, fixture: a set of stretch fixtures
13. Host size: 450 × 4 4 4 50 × 1420mm (length × width × height)
14. Host weight: (about) 45kg
15. The rated voltage voltage : AC ~ 220V50Hz