Standard:> GB 19083-2010 medical protective mask technical requirements
It is the special test equipment mainly used for testing medical masks flame retardant properties.
The device is a mask with a head mold at a certain line speed after contact with the flame, then test the burning performance on masks.
1, PLC + touch screen control system; make test more convenient operating, visualization;
2, mask fixture for the metal body head mold, can fully simulate the actual use of the state mask.
3, the burner height can be adjustable
4, automatic timing positioning made by the burner
5, burning time, fire-retardant time are automatically recorded, show the figures.
6, with a flame temperature measurement probe, flame temperature display.
7, automatic control system, automatic ignition;
8, the burner with propane (liquefied gas) and other combustible gas interface.
9, automatic timing, automatic storage of test data; easy access to experimental reports
1, head mold automatic movement line with the speed of (60 plusmn; 5) mm / s
2, the height of the flame can be adjusted to 40 plusmn; 4mm
3, with a flame temperature measurement detection thermocouple
4, the flame from the top of the burner height is(20 plusmn; 2) mm, the temperature is (800 plusmn; 50) ℃
5. The distance between the top of the burner and the lowest point of the mask is (20 plusmn; 2) mm
6, the burner height adjustment range of 30mm
7, metal head mold wall thickness of 2 ~ 3mm, resistant to flame burning
8, continued burning time and smoldering time timer: 0 ~ 9999.9 seconds, accuracy: plusmn; 0.1 seconds
9, the power supply: AC220V, 50Hz, 50W
10, Weight: about 30kg
Get a Quote
Your email address will not be published. Required fields are marked *