BS 5811/8479, IWSTM 152, NEXT 19, Mamp;S P18/P18A/P18B/P21A, IOS 4802.3, BS EN ISO
A universal pilling and snagging tester drives system to rapidly repeat pilling or snagging on fabrics to generate normal wear on samples in a fraction of the time.
Provided with unique controller allowing selection between standard and several optional speed settings.
Programmable reversing function at 30 rpm and test cycle definition facilities.
1.Equipped with four pilling Boxes;
2.With the function of forward and reverse running;
3.Optional speeds of 20, 30, 40, 45, 50, 60, 65 and 70 rpm can be chosen arbitrarily;
4.All test parameters are displayed on the LCD delete;
5.Alarm reminding as test ended;
6.Well sealed positions;
7.Automatic pause and caution to safeguard the motor.
2.4 Pilling Boxes;
3.2 Snagging Boxes;
4.A rapid mounting Jig;
5.4 sets of Cork Liners (set of 6);
6.4 sets of Snagging Points (set of 6);
7.30 Sample Tubes;
8.A Set of pilling photographs.
ICI Pilling Box Test Method：>
1、Pilling test piece preparation:
Four test pieces are cut from the fabric of size 125mm*125mm, each of them should be marked the reverse side and lengthways side. We need to test both of sides of the sample when the front and back side is not obvious. In addition, we need to cut the same size piece from fabric as compared with the test piece.
2、Pilling test procedure:
Fold two test pieces inward lengthways from the front, and then sew it for 12mm from the edge. To be a sample cell, the stitch density decides the uniform seam. Other two test pieces repeat that, but fold them inward horizontally.
Turn the inside out and turn the obverse side of the test piece into the outside of the sample cell, we cut 6mm from both ends of the test piece. Putting the sample cell on polyurethane tube, fix and tape both ends of the sample with the same distance to both edges of the tube by PVC tape.
Until any chopped fiber or matter which influence the test result out of the box, we need to clean the ICI pilling
Putting 4 prepared polyurethane tube into the pilling box and Closing the lid.
Preset the revolution by the counter and start the instrument. (The revolution of woolen fabrics: 7200r; The revolution of worsted fabrics: 14400r)
After the counter reach the revolution which is set, we take out the sample, remove the thread and unfold the test piece.
3、Pilling test result:
To assess the test piece according to the visual description. If the test piece stands between two rating, which should be recorded 1/2 rating.
The teat result is the average of all test piece rating, if the average is not an integer, the number should be modified about 1/2 rating to the nearest rating.
Get a Quote
Your email address will not be published. Required fields are marked *