The Acid-Base Concentration Meter is an on-line microprocessor based water quality monitor. The instrument is widely used in thermal energy, chemical industry, steel pickling and other industries, such as power station regeneration of ion exchange resins, chemical industry processes, etc., to continuously detect the concentration of chemical acid or alkali and to control aqueous solutions.
Features of the acid-base concentration meter:
The non-contact detection method basically eliminates corrosion, pollution and polarization effects
Using variable frequency excitation power supply and resonant circuit, and its unique signal sampling method, the detection transformer output voltage is increased by 100 times instead of single frequency excitation method.
The flow sensor has a low temtemperature magnetic structure and a ferromagnetic shield with no zero-drift phenomenon, which greatly improves the accuracy, sensitivity and stability of the instrument.
Using a microcontroller (MCU) and a dedicated interface integrated circuit, the output voltage of the detection transformer and the output current of the temperature sensor are fully digitally sampled, calculated and displayed. High-precision temperature can be achieved. Local display and remote transmission of compensation and concentration values.
Imported all-plastic sealed case, which can be used in humid, dusty and corrosive environments for a long time;
(4~20)mA standard signal output , which can be easily connected to various display and setting devices and computer systems, and users can also choose the output mode (0~10)mA or (0~20)mA.
Functional principle of theAcid-base concentration meter:
Measurement (concentration, electrode material). consists of platinum), in order to avoid electrode polarization, the meter generates a highly stable sine wave signal, which is added to the electrode, the current flowing through the electrode is proportional to the concentration of the solution to be measured, and is measured by the preamplifier, the current of the electrode is in converted into a voltage signal, and the voltage signal reflecting the concentration value is obtained after program-controlled amplification, phase-sensitive detection and filtering;
The microprocessor switches the temperature signal and concentration. The signal is alternately sampled and converted after calculation and temperature compensation calculation, and displayed as measured concentration value and real-time temperature value at 25°C.
Principle of automatic temperature compensation:
The concentration of the measured solution is non-linearly proportional to its conductivity, and the conductivity of the solution is affected by temperature. , It needs to perform temperature compensation, the temperature characteristics of different solutions are not the same, and the corresponding processing by the microprocessor becomes fast and accurate, realizing the function of automatic temperature compensation.