Some problems occasionally occur when using the universal electronic testing machine The following are some problems that may occur when using the universal electronic testing machine before If you contact the manufacturer, you can solve the problem with the following methods remedy.
1. Electronic Universal The tester's main power indicator does not light up and the device cannot move up and down
Troubleshooting method: 1. Check whether the power line connected to the testing machine is properly connected; 2. Check if theemergency stop button is pressed. If so, just flip the emergency stop button; If the fuse is blown, please take out the spare fuse and install it.
2. The host power indicator of the electronic universal testing machine is on, but the device still cannot move up and down
Troubleshooting method : 1. Start the testing machine after 15-30 seconds because it takes some time for the host system to start 2. Check whether the upper and lower limit valves are fixed in the correct position, and make sure that the moving beam does not touch the upper limit valve or the lower limit valve and there is enough clearance between the upper and lower limit valves 3. Check,whether the power supply voltage connected to the testing machine is normal. If the voltage is too low, the testing machine will not start voltage source.
3. After the computer software is online, a prompt window appears indicating overload
Troubleshooting method: 1. Check whether the communication line between the computer and the test machine is loose. If loose, please plug in and try again 2. check whether the online selection sensor is selected correctly 3. check whether the sensor has been bumped in the last test or keyboard operation 4. check whether the calibration or calibration function the software works before the problem occurs; 5. Check if the calibration value has been changed manually,Calibration values or other information in the hardware parameters.
4. Computer After the software is online, it occasionally reports that the limit information is exceeded
Remedy: 1 1. Check whether the distance between the limit rod and the moving beam is normal; 2. Open the front shell, and adjust the limit spring inside the machine along the position of the limit rod to determine whether it is stuck or invalid, to eliminate the poor contact caused by the spring problem.
Reminder: If the error of the electronic universal testing machine cannot be repaired by yourself At this time, thinkure not to disassemble the machine, so as not to damage the electronic components of the testing machine, and the manufacturer does not provide any warranty at this time. Find the manufacturer's customer service phone number through the original purchase contract of the testing machine or other things with the Information from the manufacturer and ask him to come to repair