How to deal with the fabric strength machine not returning to its original position during use>
Fabric Strength Machine sometimes has various problems during use B. The moveable crossbar cannot be reset and gets stuck. I'm worried about my operating errors, so I don't dare move it. Therefore, the editor lists several situations below and you can try to solve the problem in a more targeted way.
1. The movable beam is located at the top or bottom part of the fabric strength tester and has touched The limit valve is touched Although the power switch and control system work normally, an overflowdisplayed on alarm
Solution: The moving beam has exceeded the travel distance and the system's self-protection function activates, causing the beam to stop working. At this time, the position of the limit valve can be adjusted so that it is slightly away from the baffle on the moving beam, and then the alarm can be manually eliminated, and the moving beam can be adjusted to the middle position of the tissue force machine.
Second, the moving bar is in the center of the fabric strength tester and does not touch it Limit valve, power switch and controls work normally, no alarm indication
Solution : You can change the power supplyStop running the tissue strength device and then restart the strength device. If the moving beam still doesn't work, you can contact the manufacturer for help.
3. The moving bar is in the middle of the fabric strength tester and does not touch the limit. The valve, power switch and control system all work normally, and there is an over-limit alarm indicator
Solution: It may be that the gap between the moving beam flapper and the limit bar is too small, causing the limit bar to touch the limit switch and an overrun alarm occurs. Adjust the distance between the movable beam shutter and the limit rod to allow the movable beam to move freely and manually clear the alarm.
4. The fabric strength machine upper and lower brackets collide together, and the power switch and control system compete for production, the system will display an alarm that the load force value is too large.
Solution:The upper and lower brackets may collide with each other due to misoperation. Together, the system's self-protection function starts. Try to clear the alarm and move the moving beam to the center of the fabric strength machine; if the alarm cannot be eliminated, try to turn off the power of the fabric strength machine, restart the strength machine and adjust the position of the moving element. If it still does not work, contact the manufacturer for someone to service skicken.
Note: If there is an issue with the impact of the top and bottom mounts, it is possible that the sensor could be damaged by excessive stress. After adjusting the position of the moving bar, it is best to measure the sensor to see if there is a large error in the test accuracy. If not, a new sensor needs to be updated.