The fiber fineness analyzer is initialized before using the system
Fiber Fineness Analyzer is an instrument specifically used to test the fiber diameter, which is usually used in the textile field to measure textiles be initialized to reduce the test error of the instrument and improve the accuracy of the instrument.
1. Device assembly
(1) Install tofirst various hardware devices and then install the test software , image capture card and its driver software installed. Common office software such as Office, PDF printers and other software can be installed by the user himself.
Reminder: After purchasing the fiber gauge, the manufacturer will send technicians come to install and debug, and explain the relevant installation steps.
Second, initial device calibration
(2) Open the test software on the computer and select the appropriate test mode, e.g. wool test, cotton fiber test, etc.
(3) When selecting a suitable lens In general, when testing fiber, the magnification is mostly 200X, and the objective lens can be selected 10X. For ultrafine fiber filament, the objective lens with higher magnification can be selected become.
(4) Click on the test software, access the image, and then paste the slide, align the objective lens and adjust the focal length until the images on both sides of the fiber image are clear are.
(5) dirAlign the measured fiber under the coverslip with the objective lens, adjust the focus length, and position the stage until the fiber image becomes clearest.
(6) Measure 600 standard wool tops according to the fineness (diameter) measurement steps, and test The more roots, the more accurate. After the measurement, record the average diameter value.
(7) Open the system error correction in the operating software, and input the measured value and actual value of the standard top, and then click correction, the computer will automatically correct the system error
Reminder: Calibrate the system error of the Fiber Fineness Analyzer, usually no more than once a year.But if you are not satisfied, you can always perform a calibration Proofreading only The fineness of the fiber can only be officially measured after the system error has been eliminated.