How to set constant temperature test chamber > prevents from leaking Question:
1. Check the test chamber water refilling time Is the frequency set too high, resulting in excessive water refilling? If so, just increase the water refilling interval and decrease the frequency to solve the problem;
2. Check whether the water level box of the water course on the right side of the test chamber is too high, causing Too much water may enter At this time, it is necessary to adjust the water level box It is recommended that when adjusting the water level box, it is best to use a percontact a professional technician before taking any action.
3. Check if the floor of the lab is level, because the slope of the wet bulb angle is also a factor that causes the water in the wet bulb to become unbalanced, causing water to drip down. If this situation exists, we must Only place the device on a level floor when using it ;
4 . Check the waterway for water inlet valve is too big, resulting in too much water when you turn off the water valve directly;