The rain test chamber / waterproof test chamber is suitable for testing the protection of automobile lamps, windshield wipers, waterproof strips, locomotive instruments and low-voltage electrical enclosures, and judging whether the product casing / sealing can guarantee the equipment after the water test or during the test And the good work performance and the technical condition of the components. At the same time, the product may be affected by immersion in water during transportation or use, which is a reference base for technical product standards. gbcTextile Testing Instruments, Testing Equipment Manufacturer
Rain Test Chamber> plays an essential role in environmental testing equipment, and our rain test chamber cleaning and maintenance is also a very important step. Let's talk about the rain test chamber below. precautionsMeasures for cleaning and maintaining the rain test chamber:
1. Cleaning rain test chamber> power distribution room
The rain test chamber is exposed to the environment for a long time, power distribution indoor units, switchboards, water circuits, etc. accumulate dust . If the air is humid, it will lead to equipment failures. In order to ensure long-term stable operation of the devices, please clean the power distribution room once a month. (Warning: When performing this operation, please turn off the main power switch first to avoid electric shock!)
Method: First turn off the main switch, open the cover of the power distribution room, and use a vacuum cleaner (or a soft brush, preferably with a vacuum cleaner) to brush the dust on the electrical part and finally cover the power distribution compartment cover and tighten the mounting screws.
Second, cleaning the outer box of the rain test chamber
The device has been exposed to the environment for a long time and dust is accumulating on the outer surface .Clean the outside surface of the equipment once.
Three, Rain Test Chamber> Precautions:
1. If you hear abnormal noise during operation, hold Please turn on the machine and check it. Restart the machine so that it will not affect the life of the device.
2. The gear mechanism must be oiled regularly, and the reduction gear must be filled with clean 20# engine oil
3. After the device has been positioned, the support frame must be leveledUseful to prevent the casters from shifting after vibration during device testing.
4 . Box shower The test box has been running for a long time. If the water spray hole is found to be clogged, the piping must be disassembled, rinsed with tap water, and reassembled.
gbcTextile Testing Instruments, Testing Equipment Manufacturer