Are you still worried about not being able to run a constant temperature and humidity test chamber? Today I will introduce you the working steps of our constant temperature and humidity test chamber. First connect the power supply (the rated power supply required by the test box), then turn on the power switch and wait a few tens of seconds. Second, set it on the controller, and you can set the high temperature to 150℃; Set the low temperature according to your desired temperature. Select the model you want. There is a security indicator on the control panel. When the buzzer sounds, there is an error alarm and a safety operation prompt on the controller. Please work with confidence. Fill SFinally, fill the water tank with water when performing high-temperature, high-humidity tests.
The following is a list of points to be noted when operating the constant temperature and humidity test chamber organized by Wuhan Guoliang Instrument for all operators!
1. Please provide a power supply within the rated voltage range for the programmable constant temperature and humidity test chamber.
2. About an electric shock, a malfunctionto avoid on or failure, please do not turn on the power supply before the installation and wiring are completed.
3. The constant temperature and humidity test chamber is a non-explosion-proof product, please do not use the constant temperature and humidity programmable chamber in the environment of flammable or explosive gas.
4. It is forbidden to disassemble, process, modify or repair the constant temperature and humidity test chamber without permission, otherwise there will be a risk of abnormal movement, electric shock or fire.
5. Enclosure Ventilation system must be kept clear to avoid failure, abnormal operation, reduced life and fire
6. If the constant temperature and humidity test box is damaged or deformed during the unpacking test, please do not. If used, immediately contact the designated service personnel for maintenance and inspection.
7. When When installing and setting up the device, make sure that dust, thread ends, iron filings, or other things do not get in, otherwise incorrect operation may occurfailures or failures.
8. Wiring must be correct and grounded. Failure to ground may result in electric shock, accident with wrong action, abnormal display, or large measurement error.
9. Regularly check the terminal screws and mounting frames of the constant temperature and humidity test chamber. Please do not use them if they are loose.
10. Power supply during device operation The input terminal cover must be installed on the terminal blockert to avoid electric shock.
11. During the operation of equipment and counters, before changing settings, signal output, starting , stopping and other operations, you should fully consider safety, wrong operation will damage the working equipment or fail.
12. Please wipe the instrument with a dry cloth, do not use alcohol, gasoline or other organic solvents, and do not splash water. If water enters the device, please stop using it immediately, otherwise it may cause electric leakage, electric shock or fire.
13. It is impossible for a device to function properly. Users can be entitled to timely maintenance services when problems occur, or quickly determine the troubleshooting method themselves according to the device's troubleshooting ability, so that the device can recover its performance and become operational as soon as possible.